Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Hewlett Packard A Synthesizer. HP A Front View. Hewlett Packard A Synthesizer (click images to enlarge). The manual is available from Didier’s. View and Download HP A calibration manual online. Synthesizer/Function Generator. A Synthesizer pdf manual download. Also for: b.
|Published (Last):||3 April 2005|
|PDF File Size:||20.87 Mb|
|ePub File Size:||4.82 Mb|
|Price:||Free* [*Free Regsitration Required]|
Thsi is not a User Manual. Approved for public release; distribution is unlimited. Items of equipment used in this procedure are referenced within the text by common name as listed in table 2.
Don’t show me this message again. By manuak ‘print’ button you will print only current page.
HP A Service Manual download, schematics, eeprom, repair info for electronics experts
Page of 23 Go. Sine wave amplitude and flatness Square Wave Amplitude and Flatness a. You can help improve this publication.
The following peculiar accessory is also required for this calibration: Comments to this Manuals Your Name. Page 20 TB 4 Enter Frequency Accuracy 4 Set oscilloscope controls to measure ramp retrace time from the 90 to 10 percent points. Power supply assembly A2 — test instrument top view.
Amplitude Modulation Distortion 9. Don’t manuaal an account? To print the manual completely, please, download it. By Order of the Secretary of the Army: Record frequency counter indication.
HP A CALIBRATION MANUAL Pdf Download.
Page 22 TB Figure 5. To be distributed in accordance with IDNrequirements for calibration procedure TB Forms, records, and reports TB a 50 Hz. This manual also for: The manufacturer’s manuals and TM were used as the prime data sources in compiling these instructions.
TB 15 Set frequency counter controls to measure frequency. Multimeter will indicate between If frequency counter does not indicate between TB Figure 5. The instructions outlined in paragraphs 6 and 7 are preparatory to the calibration process. Calibration Process TB 6.
Paragraph Page Final procedure TB 4 Enter Equipment Requirements TB Test instrument parameters Amplitude accuracy with offset attenuation manal 1 into 50 load Attenuator accuracy these errors are additive with the amplitude accuracy errors Amplitude output Amplitude option high voltage output Dc offset Dc offset option Dc plus ac Not verified below 50 Hz.
If you find any mistakes or if you know of a way to improve the procedure, please let us know. Page 8 TB a 50 Hz.
Enter text from picture: